LIFEPAK Service Manual CD-Rom-Service Manual. Physio-Control LifePak AED Service Manual. Physio-Control – LifePak AED by Physio-Control. Physio-Control – LifePak AED. Download PDF. The Physio Control Lifepak AED has been designed to be used by first responders for cardiac arrest Physio-Control LifePak AED Service Manual.
|Published (Last):||2 September 2011|
|PDF File Size:||5.43 Mb|
|ePub File Size:||6.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
During the self-test, the AED displays the following messages: This could include a location servkce existing emergency equipment, such as fire extinguishers and first-aid kits.
Medtronic LIFEPAK 500 Operating Instructions Manual
Page 42 You will hear lufepak corresponding voice prompt. The Lifepak automated external defibrillator wieghs only 7 lbs and is very compact. Page 77 two hours minimum at the operating temperature prior to operation. No entries found Minor Fault Log: Forums Documents and Manuals News.
This 5500 air space or gaps under the electrodes and promotes good skin contact. If recycling is not possible, contact a Medtronic Physio-Control representative for information or assistance. The defibrillator should be easy to reach. Add to My Bench.
Children as well as adults may fall victim to sudden cardiac arrest SCA. Page 51 Equipment connections for data transfer by modem Figure 5 Turn on the modem. Page 45 20 seconds, analysis will continue. Recycling Rechargeable Batteries Recycle SLA battery paks locally according to national, regional, and local governmental regulations. Page 69 The SLA battery pak should be recharged monthly or after each use.
Physio-Control LifePak 500 Service Manual 26500-000036
Contact authorized service personnel for repair. Battery Options A rechargeable sealed lead-acid battery or one of two non-rechargeable lithium batteries sulfur dioxide or manganese dioxide provide power to the AED.
Swrvice the automatic self test detects a low battery condition or a condition that requires service, the monophasic AED activates an alarm; AEDs with a readiness display will change the indicators on the display and will not activate an alarm unless is configured ON.
My Bench Order History Sign out. Automated daily self-tests and long-lasting batteries help ensure that only minimal maintenance is required.
Page 79 J biphasic shocks will in some cases enable earlier termination of VF. The dial string may include up to 20 characters as described in Table Simple 2 or 3-button operation takes a responder through the defibrillation process with audio and visual prompts.
Page 23 Steady display indicates the AED requires service by authorized indicator service personnel; flashing indicates service is required immediately.
What could be the issue? Don’t show me this message again.
MEDTRONIC PHYSIO-CONTROL LIFEPAK | PHYSIO-CONTROL LIFEPAK Accessories
This feature tests the most important circuitry in the device to give the user a high degree of confidence that the AED is manuall for use.
After testing with Defib analyzer the indicator remains on,do you need special software to flash the unit?