The KODAK Intraoral X-ray system is equipped with a thermal safety system that prevents the generator from overheating in case of intensive use. Product, Model Kodak Intraoral X-Ray System, Catalog Numbers , , , XR — Carestream Health, Inc. Made in France Kodak Intraoral X-Ray System Performance. Operational ease. Safety. The Kodak intraoral x-ray system is designed for performance and safety.
|Published (Last):||22 August 2009|
|PDF File Size:||1.98 Mb|
|ePub File Size:||1.30 Mb|
|Price:||Free* [*Free Regsitration Required]|
My presentations Profile Feedback Log out. December 13, New law seeks to improve access to oral health care for underserved patients.
Systemm, after each exposure, the dose level is displayed for simple dose monitoring. This advanced control helps you obtain optimized images that are just the way you want them—giving you total control over your settings. Complete control The Kodak system is equipped with adjustable tension 60 or 70 kV to accentuate contrast and levels of gray shading according to your diagnostic needs. Icons are clearer and easier-to-understand so you can make the selection with just one glance.
The Kodak trademark and trade dress are used under license from Kodak. Systwm with a focal spot of 0.
Could you please send it to msobota gmx. Live Webinar Stay out of Jail: And with multiple installation configurations available, you can easily adapt the unit to match the needs of your practice. You mention that person needs a K control panel?
I did the own PBC board that had pcs of 1Mohm smd resistors in the form of a snake and 13 pcd of 1kOhm smd resistors were connected in series. Excuse me here is Amir Tabatabaei again. Advertise Increase Your Brand Awareness. My panel is without rotating timer knob. It will give you all error codes with counters. The future of periodontal therapy. The latest KODAK intraoral dental X-ray system is designed to deliver new ease kodwk use and dose management capabilities.
Please Mr Faustino can you send me the service manual and im interessed to inyour spare parts can you x–ray me suissedental gmail. Unless Trophy designed it with certain service lifetime of each unit. Multiple configurations to fit any practice Compatible with both film and digital imaging applications, the Kodak system works the way you do. Open the catalog to page 2. Open the catalog to page 1.
To make this website work, we log user data and share it with processors. There are two grooves on the plastic cup that must match those in the bottom of the housing. Share buttons are a little bit lower.
Used & Refurbished Dental Equipment
Hello My Kodak is always giving the message of KV error. Hello,was there any sparks inside the tubehead untill now? On the power board set the dipswitch 4 of SW2 to ON Switch “ON” the board while maintaining the X-ray switch of the control panel activated until the end of the self-calibration sequence.
Then insert the assembly gently inside the metal housing, taking care with the position. Continuous Flow and Accurate Measurement. Can be located anywhere from close to the tube to outside of the operating room Timer can also be mounted on the wall-framework.
Other information will follow by mail. As a part of its mission to always provide dentists with only the best solutions for improved patient care, ContacEZ continues to refine, improve and expand its product lines to make them safer and more effective than effort.
You need the K control panel to service the unit. Ad Statistics Times Displayed: Can anybody send me service syystem or some instructions how to calibrate KV. A short beep sequence is kpdak at the end if the self-calibrations succeed. In addition, the system emits constant radiation to maximize the amount of useful x-rays and significantly reduce the levels of radiation.
Kodak Kv/error – Service Technicians Forum
About project SlidePlayer Terms of Service. It gives KV Error along with bip and on the control panel two green lights turn on. Would be thankful if you could send me the tech manual as well. Sometimes the plastic cup bends itself so it pulls up the ceramic resistor. I drill a 1 mm hole near anode’s contact in the plastic shell of the tube to be sure that all air bubbles are eliminated.
May 11, Business The best solution for IPR just got better As a part of its mission to always provide dentists with only the best solutions for improved patient care, ContacEZ continues to refine, improve and expand its product lines to make them safer and more effective than effort.
With the new system timer, dental professionals can quickly and easily select the most appropriate exposure settings based on the teeth to be examined, the patient size and receptor used.
The voltage devider must be immersed in oil completely, no air bubbles present in oil! Avoid Coding Errors 12 Feb Live Webinar 19 Mar Auth with social network: Related Searches Flat panel detector Dental imaging software Medical radiology software Cephalometric analysis software Dental software Simulation software Analysis software.